Maximum Input Voltage: CAT I and CAT II: 300VRMS (10×), Installation Category; CAT III: 150VRMS (1×); Installation Category II: derate at 20dB/decade above 100kHz to 13V Pea'k AC at 3MHz* and above. For non-sinusoidal waveforms, Pea'k value must be less than 450V. Excursion above 300V should be of less than 100ms duration. RMS signal level including all DC components removed through AC coupling must be limited to 300V. If these values are exceeded, damage to the oscilloscope may occur.
Sample Rate Range: 500MS/s--1GS/s
Waveform Interpolation: (sin x)/x
R'ecord Length: 40'K
Sample Rate and Delay Time Accuracy: ±50ppm(at over any ≥1ms time interval)
Vertical Resolution: 8-bit resolution, all channel sampled simultaneously
Position Range: 2mV/div to 200mV/div, ±2V 200mV/div to 5V/div, ±50V
Analog Bandwidth in Normal and Average modes at BNC or with probe, DC Coupled: 2mV/div to 20mV/div, ±400mV; 50mV/div to 200mV/div, ±2V 500mV/div to 2V/div, ±40V; 5V/div, ±50V
Math: +, -, *, /, FFT
FFT: Windows: Hanning, Flatop, Rectamgular, Bartlett, Blackman; 1024 sample point
Bandwidth Limit: 20MHz
Low Frequency Response (-3db): ≤10Hz at BNC
DC Gain Accuracy: ±3% for Normal or Average acquisition mode, 5V/div to 10mV/div; ±4% for Normal or Average acquisition mode, 5mV/div to 2mV/div
DC Measurement Accuracy, Average Acquisition Mode: When vertical displacement is zero, and N ≥16:± (3% × reading + 0.1div + 1mV) only 10mV/div or greater is selected; When vertical displacement is not zero, and N≥16: ± [3% × (reading + vertical position) + 1% of vertical position + 0.2div]; Add 2mV for settings from 2mV/div to 200mV/div; add 50mV for settings from 200mV/div to 5V/div
Volts Measurement Repeatability, Average Acquisition Mode: D1elta volts between any two averages of ≥16 waveforms acquired under same setup and ambient conditions
Trigger Types: Edge, Video, Pulse, Slope, Over time, Alternative
Trigger Sensitivity (Edge Trigger Type): DC(CH1,CH2): 1div from DC to 10MHz; 1.5div from 10MHz to 100MHz; 2div from 100MHz to Full; DC(EXT): 200mV from DC to 100MHz; 350mV from 100MHz to 200MHz; DC(EXT/5): 1V from DC to 100MHz;1.75V from 100MHz to 200MHz; AC: Attenuates signals below 10Hz; HF Reject: Attenuates signals above 80kHz; LF Reject: Same as the DC-coupled limits for frequencies above 150kHz; attenuates signals below 150kHz
Trigger Level Range: CH1/CH2: ±8 divisions from center of screen; EXT: ±1.2V; EXT/5:±6V
Trigger Level Accuracy( typical)Accuracy is for signals having rise and fall times ≥20ns: CH1/CH2: 0.2div × volts/div within ±4 divisions from center of screen; EXT: ± (6% of setting + 40mV); EXT/5: ± (6% of setting + 200mV);
Set Level to 50%(typical): Operates with input signals ≥50Hz
Video Trigger Type: CH1, CH2: Pea'k-to-Pea'k amplitude of 2 divisions; EXT: 400mV; EXT/5: 2V
Signal Formats and Field Rates, Video Trigger Type: Supports NTSC, PAL and SECAM broadcast systems for any field or any line
Holdoff Range: 100ns ~ 10s
Pulse Width Trigger:
Pulse Width Trigger Mode: Trigger when (< , >, = , or ≠); Positive pulse or Negative pulse
Pulse Width Trigger Point: Equal: The oscilloscope triggers when the trailing edge of the pulse crosses the trigger level. Not Equal: If the pulse is narrower than the specified width, the trigger point is the trailing edge. Otherwise, the oscilloscope triggers when a pulse continues longer than the time specified as the Pulse Width. Less than: The trigger point is the trailing edge. Greater than (also called overtime trigger): The oscilloscope triggers when a pulse continues longer than the time specified as the Pulse Width
Pulse Width Range: 20ns ~ 10s
Slope Trigger Mode: Trigger when (< , > , = , or ≠ ); Positive slope or Negative slope
Slope Trigger Point: Equal: The oscilloscope triggers when the waveform slope is equal to the set slope. Not Equal: The oscilloscope triggers when the waveform slope is not equal to the set slope. Less than: The oscilloscope triggers when the waveform slope is less than the set slope. Greater than: The oscilloscope triggers when the waveform slope is greater than the set slope.
Time Range: 20ns ~ 10s
Over Time Mode: Rising edge or Falling edge
Time Range: 20ns ~ 10s
Trigger on CH1: Internal Trigger: Edge, Pulse Width, Video, Slope
Trigger on CH2: Internal Trigger: Edge, Pulse Width, Video, Slope
Readout Resolution: 6 digits
Accuracy (typical): ±30ppm (including all frequency reference errors and ±1 count errors)
Frequency Range: AC coupled, from 4Hz Min to rated bandwidth
Signal Source: Pulse Width or Edge Trigger modes: all available trigger sources The Frequency Counter measures trigger source at all times, including when the oscilloscope acquisition pauses due to changes in the run status, or acquisition of a single shot event has completed. Pulse Width Trigger mode: The oscilloscope counts pulses of significant amplitude inside the 1s measurement window that qualify as triggerable events, such as narrow pulses in a PWM pulse train if set to < mode and the width is set to a relatively small time. Edge Trigger mode: The oscilloscope counts all edges of sufficient amplitude and correct polarity. Video Trigger mode: The Frequency Counter does not work.
Cursor Measurement: Voltage difference between cursors: ΔV Time difference between cursors: ΔT Reciprocal of ΔT in Hertz (1/ΔT)
1 x Digital Storage Oscilloscope 2 x Probes (PP150) 1 x Oscilloscope Power Cable 1 x USB line
1 x Hantek data CD 1 x Manufacturer Certificate
Payment & Security
Your payment information is processed securely. We do not store credit card details nor have access to your credit card information.
Our policy lasts 30 days. If 30 days have gone by since your purchase, unfortunately we can’t offer you a refund or exchange.
To be eligible for a return, your item must be unused and in the same condition that you received it. It must also be in the original packaging.
Several types of goods are exempt from being returned. Perishable goods such as food, flowers, newspapers or magazines cannot be returned. We also do not accept products that are intimate or sanitary goods, hazardous materials, or flammable liquids or gases.
Additional non-returnable items:
Downloadable software products
Some health and personal care items
To complete your return, we require a receipt or proof of purchase.
Please do not send your purchase back to the manufacturer.
There are certain situations where only partial refunds are granted (if applicable)
Book with obvious signs of use
CD, DVD, VHS tape, software, video game, cassette tape, or vinyl record that has been opened
Any item not in its original condition, is damaged or missing parts for reasons not due to our error
Any item that is returned more than 30 days after delivery
Refunds (if applicable)
Once your return is received and inspected, we will send you an email to notify you that we have received your returned item. We will also notify you of the approval or rejection of your refund.
If you are approved, then your refund will be processed, and a credit will automatically be applied to your credit card or original method of payment, within a certain amount of days.
Late or missing refunds (if applicable)
If you haven’t received a refund yet, first check your bank account again.
Then contact your credit card company, it may take some time before your refund is officially posted.
Next contact your bank. There is often some processing time before a refund is posted.
If you’ve done all of this and you still have not received your refund yet, please contact us at firstname.lastname@example.org.
Sale items (if applicable)
Only regular priced items may be refunded, unfortunately sale items cannot be refunded.
Exchanges (if applicable)
We only replace items if they are defective or damaged. If you need to exchange it for the same item, send us an email at email@example.com and send your item to: 9955 s May Chicago Illinois US 60643.
If the item was marked as a gift when purchased and shipped directly to you, you’ll receive a gift credit for the value of your return. Once the returned item is received, a gift certificate will be mailed to you.
If the item wasn’t marked as a gift when purchased, or the gift giver had the order shipped to themselves to give to you later, we will send a refund to the gift giver and he will find out about your return.
To return your product, you should mail your product to: 9955 s May Chicago Illinois US 60643
You will be responsible for paying for your own shipping costs for returning your item. Shipping costs are non-refundable. If you receive a refund, the cost of return shipping will be deducted from your refund.
Depending on where you live, the time it may take for your exchanged product to reach you, may vary.
If you are shipping an item over $75, you should consider using a trackable shipping service or purchasing shipping insurance. We don’t guarantee that we will receive your returned item.
You may also like
FAST Delivery Worldwide
small shipping fee's
Satisfied or refunded
Refunds are accepted with-in 30 days. More than 30 days with some exceptions.
FOR SALES QUESTIONS
All payments are SECURE and verified against Credit Card Fraud.